Payment & Shipping Terms:
|Autoclave Sterilizer Machine
|Single Or Double Horizontal Sliding Door
|Chamber Size Mm:
|Outer Size Mm:
|Loading Trolley Quantity:
|With 2 Unit At Least
|Transfer Trolley Quantity:
|Single Door 1 Unit,double Door 2 Units
|Machine And Accessories Used In Production Of Biopharmaceutical Factory
Steam Autoclave Sterilizer,
Unique non-stick door gasket
Specially formulated to eliminate sticking and to avoid door jamming,FDA food grade silicon.
Minimize steam exposure and extend working life.
Whisper-quiet air compressor optional
Minimize noise,eliminate need for air supply,supply constant air pressure to gasket
Avoids steam exposure to electronics and extends touchscreen life
Counterbalanced sliding door
Simplifies one button,lifts by motor
One style solenoid valve used throughout sterilizer
Simplifies maintenance,non-proprietary valve reduces cost
Integrated post-vac drying cycle
Vacuum drying phase eliminates steam and removes moisture from the load((i.e. reduces wet-packs)
Pressure gauge calibration
Permits periodic calibration to maintain accuracy,extend component life
All sliding door sterilizers are custom made to exact configuration,functionality and options
Automatic power door operation
Powered door operation allows 1 finger open/close operation.The door can be manually operated in the event of power loss.
The Autoclave Sterilizer Machine shall have minimum of the following alarms and safety features, but not limited to:
High Temperature over shoot alarm
Too long time to create vacuum
Vacuum pump trip alarm
Utility failure alarm
All the motors shall be provided with overload relay.
During the sterilization period if the chamber temperature falls below the set temperature,
the sterile hold period counting will stop. The counter shall restart again once the set
temperature is reached.
This autoclave sterilizer machine is horizontal with sliding horizontal opening double door,
horizontal chamber type (rectangular chamber with smooth round corners and jacket)
with fully automated operating system.The entire construction including all components
shall be fully compliant with cGMP requirements.
A stainless steel flush mounting shall be fitted between the sterilization chambers and the
sterile area wall to avoid any cross flow of air between the sterile and non-sterile side.
3. Vacuum pump
The sterilizer equipped with a water-ring type vacuum pump.
4. Steam condenser
Vendor shall provide a steam condenser made of AISI 304 fitted in the exhaust/vacuum line to
condense the steam before entering the vacuum pump.
The chamber equipped with two validation ports with dummy & adaptor for insertion of
multiple temperature sensors (minimum upto 16 temperature sensors).
The equipment has a provision on the steam feeding pipe which allows the installation of
an appropriate device (Client’s scope) for the execution of steam quality tests (non condensable
gas test, superheated gas test and dryness test).
6.Air Intake Filter
The sterilizer has a sterilizing grade hydrophobic air intake filter of 0.22micron pore size for
vacuum break. The air intake filter shall be placed on the sterile side of the sterilizer. Option shall
be provided to sterilize the filter and perform integrity check. Relevant RTD for process control
and monitoring inside the filter housing shall be provided.
The Autoclave sterilizer Machine shall be suitable for handling the following materials:
Containers and metal materials
Machine and accessories used in production
Basic biopharmaceutical products which can withstand heat treatment at high temperatures.
Clean room Garments
Silicon braided tubes etc.
Minimum sterilization requirements selectable as minimum F0 target.
Sterilization control selectable on time basis (with or without F(T,Z) calculation) or F(T,Z) basis.
Contact Person: Ms. JULIET